Agilent. EA – EA. Programmable DC Power Supplies. Data Sheet. View at Test Equipment Depot | | Find great deals for Agilent EA DC Power Supply 30 Day. Shop with confidence on eBay!. On Sale Agilent EA. We take purchase orders and can provide special school, government and corporate pricing.
|Published (Last):||4 July 2017|
|PDF File Size:||17.94 Mb|
|ePub File Size:||6.80 Mb|
|Price:||Free* [*Free Regsitration Required]|
Accredited Calibration with Data. Transcat makes it easy: This is actually towards the front of the power supply making it a little awkward to pick up.
The underside of the board has some giant copper traces, and a large portion has the agllent removed and has been tinned. In my case the encoder would miss steps or sometimes step in the wrong direction. Keysight BenchVue software for the PC eliminates many of the issues around bench testing. Skip to content I recently bought a used Agilent power supply e3640z a very sad condition. I recently bought a used Agilent power supply in a very sad condition. After cleaning the grime off the front panel and removing the marker from the case I e364a0 the E into running a self test which it passed without hesitation.
Ships in weeks.
Keysight Technologies E3640A
I hope my e3640aa can help someone else out if they are having similar issues with their e360a. There are three clips along the top which can be accessed from the inside. After doing a little research I found that it was a e3640w issue for the rotary encoders in these power supplies to fail. After the knob is pulled off the front the dark cover needs to be removed.
Request a Rental Quote. The new encoder worked perfectly and made the supply work like new. I cleaned up the flux left over with a little rubbing e3640s and put everything back together.
Request a Calibration Quote Transcat can calibrate your existing instruments. Because the knob is the main method of user input e36640a than RS or HPIBit made the power supply almost impossible to use. By making it simple to connect, control agilnet, and automate test sequences you can quickly move past the test development phase and access results faster than ever before with just a few clicks. I made sure the holes had minimal solder before pressing the new encoder into place and soldering.
Other than that there are a lot of passive components and a MOV soldered on near the output. Leave a Reply Cancel reply. Don’t forget the Software! Rent From Transcat Do you need a test instrument for a short time? If you forget to remove these the PCB on the other side will not be able to be removed. Now to actually get it installed.
Agilent EA Power Supply – File Exchange – MATLAB Central
Accredited Calibration without Data. New Purchase Calibration Learn More Save time and money by calibrating your new instrument before it leaves our warehouse, ensuring it arrives ready e3604a be put into service. This product is available for shipping to United States and Canada only. Finally the board is out and the rotary encoder can be replaced!
Since the repair I have been using the power supply for a little over a week and I have had no issues whatsoever, and I am very happy how it turned out. Fortunately Agilent has an online shop that supplies many wear-and-tear items for their various products. The over voltage features as well as the programmability of the supply both checkout out as well. After it is loose the cover tilts agillent and is easily removed.
Finally on the other side where the PCB is the tab routed in the board needs to be lifted gently with a small screwdriver until it clears the white plastic tab. It does have two speeds loud-ish and louder? The front panel is easily separated from the main body of the power supply.
After removing a couple of screws the whole inside of the power supply slides out the front. The power supply started up perfectly the first time and I made sure to run the self test again just in case. I also made sure to order a new knob to replace to old grungy one to complete my work on the supply.
Request a Calibration Quote. Would you like this item calibrated? It is also the same encoder used in many different Agilent products. The whole board will then slide over to the right in this picture and is easily pulled out.